The Healforce Biosafety HF Safe AC1200 Cabinet is defined as a ventilated cabinet for personnel, product and environment protection and is suitable for use with agents that require Biosafety Level 1, 2 or 3 containment.
Healforce HF Safe AC1200 is designed to meet the operating principles and performance aiteria:
US Standard ANSI/NSF49:2002
European Standard EN12469:2000
Chinese Standard YY0569-2005
-ULPA Filter (Optional for Manual Window Series)
99.999% efficient (0.1-0.2µm DOP) ULPA filter Is constructed of pleated borosilicate glass fibers.
-HEPA Filter (Standard for Manual Window Series)
99.999% efficient (0.3µm DOP) HEPA filter is constructed of pleated borosilcata glass fibers.
-Filter Life Indicator
The pressure value is displayed on the control panel to show the life utility of the main filter and exhaust filter.
-A microprocessor oontrol system,, together with LED display makes it clear for all control and safety functions. lntelligent diagnostics of problems.
-True airflow velocity (both downflow and innflow) sensing technology, with temperature compensaUon for improved sensor accuracy. The velocity value continuous digital display on the front LED for constant monitoring.
-UV timer function to control the decontamination cycle and maximize lamp life.
-Easy to Decontaminate and Disinfect
The working chamber of the safety cabinet is easy to be cleaned and disinfected. The stainless steel counter surface can be took out for autoclaving.
-Weight-balanoed Front Window
-70% air recirculation
-30% air exhaust
-Maintain a minimum average inflow velocity of 0.53m/s
-Have all biologically contaminated ducts end plenums under negative pressure or surrounded by negative pressure ducts and plenums
-Cabinet Leak Test
Heal Force cabinets have to pass pressure decay test as defined
in the European Standard EN12469:2000, US Standard ANSI/NSF49:2002
and Chinese Standard YY0569- 2005.
-Kl-Discos Containment Test
Heal Force cabinets are tested using the
Kl-Discus method according to European
Standard EN12469:2000 and Chinese
Standard YY0569-2005 for containment and
operator safety at the manufacturing site.
-FIiter Leak Test
Heal Force cabinets are tested using an aerosol photometer according to
European Standard EN12469:2000, US Standard ANSVNSF49:2002 and
Chinese Standard YY0569-2005. This test detennlnes the integrity of
downflow and exhaust filters, filter housings and fitter mounting frames.
-Automatic sett-filtering cycle at start-up requires a fixed period of warm-up period
during which the cabinet workzone Is purged of contaminants before use.
-Intrinsically-safe negative pressure design.
-The motor/blower system is able to compensate automatically to maintain the downflow air at the speed of 0.35m/s+-0.025 without interruption to guarentee the safe performance.
-Interlocked UV light only operates when blower end fluorescent light are off and sash is dosed.
-Front window glass is toughened glass (standard) or non-reflecting multi-layer safety glass (optional), which provides enhanced UV protection for operators.
-Audible and visual alarms prompt the operator in case of any unsafe condition.